Welcome to our second installment of Instrument of the Month. For September, we’re highlighting Baymax!
What is “Baymax”?
Baymax is a Zeiss AxioScan.Z1 with automated slide scanning capabilities.
Why the name Baymax?
All MIC instruments have traditionally been nicknamed with sci-fi/robot references in mind. Due to its round, simple appearance, we thought Baymax (from Big Hero 6) was an appropriate name. We hoped it would be as friendly and helpful as its namesake!
What imaging capabilities does Baymax have?
Baymax is equipped with a Hitachi 3-color camera for brightfield imaging and a Hamamatsu Orca Flash 4.0 camera for epifluorescence imaging. It has filters for DAPI (Zeiss Filter Set 49), eGFP (FS 38 HE), Cy3 (FS 43 HE), mPlum (FS 64 HE), and Cy5 (FS 50). We have 25 trays, and each tray holds 4 slides (standard 1″ x 3″), for a total of 100 slides.
What objectives does Baymax have?
Baymax has 3 objectives (all air): a Fluar 5X/0.25 NA, a Plan-Apo 10X/0.45 NA, and a Plan-Apo 20X/0.8 NA.
How does slide loading and detection work?
Slides are loaded in holders, which are then loaded into trays. The trays make up a column, which is referred to as the magazine. If you’re of a certain age, the best analogy to use is old-school CD trays. Once the trays are loaded and the door is closed, then the scanner does a quick check of which trays are holding slides.
How does slide scanning work?
The preview camera takes a snapshot of the label and the slide. (See below for an example of the preview.) Then, the tissue will be either manually or automatically detected, depending on the level of contrast in your tissue. Using a personalized scanning protocol, the scanner determines the best focal plane, then acquires the images in a tiled-fashion. The scanning software will then stitch the tiles together.
What goes on behind that big white box?
Here’s a video of the tray unloading and loading. Notice the magazine moving up and down on the right side of the frame. (Video credit: Mo Kaze.)
Here’s a video of the coarse and fine focus. Notice the turret changing from the 5X objective (coarse focus) to the 20X objective (fine focus). (Video credit: Mo Kaze.)
What kind of projects are most suitable for Baymax?
Slides that are all prepped very uniformly (e.g., thickness, contrast/signal) are great for automated slide scanning. We’ve had success with histological sections of tissues. Fixed samples only – slides should be clean and dry.
How long does each slide take to scan?
It varies widely, depending on the size of your sample, the magnification you’re using, and the overall exposure time (i.e., 4-channel fluorescence will always take longer than brightfield). The good news is that it’s automated, so once you set up your scan, you can press “start scan” and come back when your slides are done. We even have remote desktop set up, so you can login remotely and check on your scan status.
Can someone help me with slide scanning?
Yes! We are now offering full-service slide scanning. Please see the information sheet for more details.
Who do I contact for more information about the microscope, training, etc.?
As always, you can email Jen or Holly for more information. The MIC training page lists the steps on how to get trained on all of our instruments. If you are interested in learning more about Baymax, come to the special November Open MIC meeting on November 12th (11:30am in LSA 347) to learn about the automated instruments available at the MIC and at QB3!